Because Dentrix Ascend Imaging is a web-based application, it is available immediately for use once you have purchased it and it is enabled by Henry Schein. However, for each computer that you want to acquire images from, you must install the following on that computer:
Drivers - The drivers for the intraoral and/or extraoral acquisition devices that you want to use. For the complete list of acquisition devices that are supported by Dentrix Ascend Imaging, see "Supported Acquisition Devices."
Acquisition agent - A small application that runs in the background and handles the communication between your acquisition devices and Dentrix Ascend. You do not need to install the acquisition agent on computers that you will be using to only view images.
Notes: During the installation of the acquisition agent, a security certificate is installed automatically to allow for secure communication between the browser running Dentrix Ascend and the acquisition agent.
Note: If you do not perform the necessary setup on a computer that you want to acquire images from before attempting to acquire images, when you attempt to begin acquiring images (see step 2 of "Acquiring Images"), you will be asked to perform the necessary setup and will not be allowed to proceed with the acquisition until the setup is complete.
To install and configure the acquisition agent (Windows only)
Mac users: To install the acquisition agent, please contact our support team at 1-855-232-9493.
Important: Prior to installing the acquisition agent, verify that the Microsoft .NET 4.0 framework is installed. If .NET 4.0 is not installed, you must install it before you attempt to install the acquisition agent.
Run Acquisition Agent Installer.exe.
On the welcome screen, click Next.
On the Choose Components screen, leave Acquisition Agent selected, and then click Next.
On the Choose Install Location screen, leave the default Destination Folder selected, or click Browse to select a different location. Then, click Install.
On the completion screen, click Finish.
If you need to change the default settings of the acquisition agent (even though the defaults should suffice), after you complete the installation, right-click the Acquisition Agent icon in the notification area of the Windows task bar, and then click Preferences.
The Preferences dialog box appears.
Set up any of the following options as needed:
Correction File Path - For Dexis sensors, this is the default location of the calibration files (C:,Dexis,Flashdir). If the calibration files are located elsewhere, click Browse to locate and select the correct folder.
Use Binned Acquisition - In the Dexis imaging software (versions 9 and 10), "binned" is the default operating mode for Dexis sensors, so this check box is selected by default for Dentrix Ascend Imaging.
Use Buffered TWAIN Transfer - This option is for TWAIN drivers and is selected by default. If there are issues with TWAIN sources delivering images properly to the acquisition agent, clear this check box to use the native transfer mode, which some TWAIN sources require or recommend.
IAM Settings - Click to configure Optime or OP30 settings.
Joy Number and Use Joystick with Generic Camera - If you use a "Generic" intraoral camera with a joystick button interface instead of proprietary buttons, select the Use Joystick with Generic Camera check box. The camera button will use a specific joystick number. This is typically zero if no other joysticks are installed on the computer. Enter a number from 0 - 10 for Joy Number.
If you use a Digital Doc camera, and it has been installed to use a port other than zero, you must specify the appropriate joystick number.
Intraoral and Extraoral TWAIN rotation - The Acquisition Agent uses settings in an .ini file to rotate and/or flip an image from TWAIN intraoral or extraoral imaging device prior to being delivered to Dentrix Ascend Imaging. You can modify these settings to correct the rotation/orientation of images from TWAIN devices. The .ini file is located in the Acquisition Agent's folder on the computer. The .ini file lists all the TWAIN devices that have been used by Ascend's imaging. A device will be listed under the category that corresponds to the type of device that it is: PanoramicTwain (for panoramic images) or DigitalSensorTwain (for intraoral images). Refer to the examples below.
Planmeca Twain Source=RotateNoneFlipNone
Next to the device name, you can change the value after the equals sign (=) to one of the following values:
Note: The acquisition agent logs actions that it performs for debugging, troubleshooting, and performance monitoring. To view the log, click Show Log Files.
To set up imaging devices
Do one of the following:
From the Settings menu, click Imaging Settings.
Use the Patient Search box to access a patient's record. From the patient's clinical chart, click the Imaging Settings option on the Imaging button menu.
The Imaging Settings page opens.
Select one of the following options:
View Only - Use this computer to view images and not acquire them. Skip step 3.
View & Acquire - Use this computer to view and acquire images. Proceed to step 3.
Click Install Acquisition Agent. Follow the instructions to install the acquisition agent and drivers for the selected acquisition devices.